Korean J Lab Med. Jun;27(3) [Evaluation of the Abbott Cell-Dyn Sapphire hematology analyzer]. [Article in Korean]. Park Y(1), Song J, Song S. Name of instrument, CELL-DYN Sapphire*. First year installed in U.S./Outside U.S./No. of units sold in , //—. No. units installed in U.S./Outside. The CELL-DYN Sapphire hematology system is designed to address routine testing needs of high-volume hematology laboratories and difficult.
|Country:||Republic of Macedonia|
|Published (Last):||6 December 2018|
|PDF File Size:||17.5 Mb|
|ePub File Size:||10.10 Mb|
|Price:||Free* [*Free Regsitration Required]|
The Cell-Dyn Sapphire flagged 7 samples, but in only 2 of those samples were metastatic cells present according to the differentiation of the cytocentrifuged preparation. Leukocyte counts in cerebrospinal fluid with the automated haematology analyzer CellDyn and the urine flow cytometer UF Diamond Diagnostics expressly disclaims any affiliation with products it does not manufacture, as well as sponsorship by other manufacturers.
sapphiee But more interestingly, a screening opportunity exists within the very good sensitivity of Table 4 summarizes the clinical performance characteristics of the NC count for the different body fluids with their respective cutoff values for normality. Cell Dyn Number of Assays: Within-run imprecision of the microscopic NC count was also investigated at low cell counts. October 6 th1 Reply Post a Reply. Performance evaluation of the application of abbptt fluids on the Sysmex XE series automated haematology analyzer.
Figure 1 shows the imprecision CV as a function of the cell count.
Products are available for international distribution only unless otherwise indicated. Close mobile search navigation Article navigation. On the other hand, the Cell-Dyn Sapphire is clinically useful for distinguishing WBC admixture from peripheral blood caused by traumatic lumbar puncture.
Abbott Diagnostics CELL-DYN SapphireÂ® | Forums | Questions | Discussions | Help | LabWrench
Comparative evaluation of the Iris iQ body fluid module with manual hemacytometer count. The positive and negative likelihood were respectively 3. Wilcoxon tests for paired samples were used because normal distribution could not be demonstrated.
In 10 samples of serous fluids, metastatic cells were ydn with microscopic NC differentiation of the cytocentrifuged preparation. Inquire Here Sample Input Type: Comparison between Cell-Dyn to Cell-Dyn Sapphire Please could you kindly explain in brief the differences between the two Sapphlre We were also able to show that it suffices to run 1 blank sample after a normal blood sample to minimize carryover for the WBC and RBC counts.
It is a multi-parameter automated hematology analyzer designed for in vitro diagnostic use in counting and characterizing blood cells.
Abbott CELL-DYN Sapphire Hematology System
Inquire Here Point Of Care: Inquire Here Reagent Type: Evaluation of carryover from normal high cell count blood samples to low cell count body fluid samples indicated that after analysis of a blood sample, at least 1 blank sample should be run before the WBC count and the RIC and ROC drop to the background level data not shown.
For assessing the differential count, the percentages of lymphocytes and monocytes determined by the Cell-Dyn Sapphire were added, and this sum was compared with the mononuclear cell MNC count found on the cytocentrifuged preparations, which included lymphocytes, monocytes, and, in some samples, epithelial cells and macrophages. However, manual hemacytometer counts could even be more imprecise at low cell concentrations. Up to sample autoloader and open vial sample handling. In the present article, we describe its applicability for the distinct body fluids, and we highlight limitations and caveats.
To assess the ability of the Cell-Dyn Sapphire to distinguish normal from abnormal body fluid samples, Bayesian statistics were used with the manual hemacytometer count as the comparison method.
Abstract The enumeration and identification of blood cells in body fluids offers important information for the diagnosis and treatment of various medical conditions. Manuals and Resources Note: The analyses were performed as soon as possible and for CSF, always within 1 hour from the time of sample collection. The names and logos of manufacturers, their instruments, and their products referred to herein may be protected by trademark or other law, and are used herein solely for purpose of reference.
[Evaluation of the Abbott Cell-Dyn Sapphire hematology analyzer].
September 25 th1 Reply Post a Reply. Experienced observers examined at least nucleated cells. As a consequence, the Cell-Dyn Sapphire had a very good sensitivity Inquire Here for a complete list! Each dilution was examined 10 times.
Full data management with celll storage for 10, cycles with graphics for easy review of patient results. All samples were collected into sterile tubes without anticoagulants. DMononuclear cells. Solid lines are the regression lines and dotted lines, the identity lines. Abbott products have helped transform the practice of medical International Council for Standardization in Haematology: Abbktt within-run imprecision of the microscopic NC count at low cell counts was comparable to that of the automated method and ranged from